Available with standard Ranger 2 systems, or those enabled with Survey, DP and Professional feature packs. DP-INS For tightly integrated acoustically-aided inertial DP operations ,you will need to install our DP-INS sensor to work alongside your Ranger 2 system. DescriptionMultiRanger is a versatile short to medium-range ultrasonic single and multi-vessel level monitor/controller for virtually any application in a wide range of industries. Don’t forget that the safest engineered level measurement solution includes switches for back-up, overfill, low level and dry run protection.
Similar to the digital process variable, the output current’s digital value should be read using a hand-held communicator. Where through-hull deployment via a gate valve is not available or practical, a through-tube machine is available. Optimised USBL The positioning accuracy obtainable from Ranger 2 can be further improved by co-locating Lodestar, our premium quality Attitude and Heading Reference system (AHRS), with your vessel’s 6G acoustic transceiver. Both units incorporate three Ring Laser Gyroscopes and three accelerometers selected for use for their performance, high mean time between failure (MTBF) and ease of export (non ITAR). These sensors have highly stable error characteristics and are compensated for temperature variation. Charger port conveniently located in the tiller, front and rear LED lighting, 9-inch wheels at front and rear, an additional 2-inch deck space and an increased weight capacity of 23 stone 3lbs.
Materials Needed for Calibrating a DP Level Transmitter Understanding the Outputs of a Transmitter Before the process of calibrating a level transmitter can begin, it is important to fully comprehend the three outputs of a level transmitter. See the Siemens MultiRanger 200 HMI for the latest technology. The MultiRanger 100/200 are still available, but a newer version has been released. Known as Optimised USBL, the advantage of this configuration is that raw USBL range and bearing data is simultaneously processed with the Lodestar’s attitude data. Lastly, the multimeter is connected in series from the transmitters negative voltage to the mA connector and from there back to the power supply.